4-23. GENERAL. (CONT)
Always keep in mind the warnings and cautions while performing the PMCS procedure or
any routine checks.
ORGANIZATIONAL PREVENTIVE MAINTENANCE CHECKS AND SERVICES (PMCS)
Visually check generator cover for chips, cracks, dents, or
Turn impeller and check for excessive friction, binding, or
Visually check vane tail for dents or opening of seams.
Visually check connector housing for chips, cracks, dents,
Visually check for defective or missing O-rings which will
Unplug indicator before performing following checks on indicator.
Check for frayed wires or loose connections on indicators
or TB201, especially terminals 1 and 9 on TB201.
Check fuse holder F201 for cracks or frayed wires.
Requisition any operator and organizational maintenance
manuals covering the AN/GMQ-11 and its components if missing
or in unusable condition.
Check DA Pam 310-1 for MWO's required for the AN/GMQ-11 and
its components. Check equipment to see if current MWO's have
been complied with and the MWO number is stamped as required.
Perform or request modification as applicable.