bined motion resulting from the movement of both sets of cells is
transmitted to the bimetal shaft from the free end of the main lever
by means of an adjustable link. From this point, the functioning of
the parts is identical to that of Barometers ML-102-E and ML-102-F.
c. This model utilizes a coil spring concentric with the pointer shaft
to take up backlash in the gear teeth instead of the flat hairspring
used in the other models.
17. BAROMETERS ML-102-E and ML-102-F (fig.
8). a. Mechanism.
(1) The mechanism of Barometers ML-102 -E and ML-102 -F is
mounted below the dial anti is built upon a brass casting which is