4-16. INDICATOR CABLING INSTALLATION. (CONT)
Power cable comes either hard wired to terminals 1 and 9 on TB201 or has plug P401 (8)
that connects to receptacle J401 (9).
6. Connect power cable plug P401 (8) to receptacle J401 (9).
4-17. INDICATOR INSTALLATION.
MATERIALSIPARTS: Nails, #3D (two required)
Wood screws, 1-inch (two required)
The indicator can be mounted on either a flat level surface or secured to a wall.
When mounting indicator on flat level surface, make sure there is enough room left all
around indicator for ease of maintenance and connection of cables.
into indicator case. Perform steps 1 and 2 to recess receptacles.
1. Remove knurled reducer (1), retainer collar (2), and adapter plug (3), and loosen nut (4).
Store items (1), (2), and (3) for future use.
2. Push receptacle (5) into indicator case enough to allow last few threads of receptacle to hold
4. Install two nails or screws in wall and hang indicator by two slotted holes on rear of indicator